Order by: Relevancy | Date
Dec 11, 2019 | Although this is not technically a bottom termination component, the method of attachment of the the
Aug 1, 2019 | hi Team, observed component skew during reflow process. Inductor location. I have attached the im
Sep 20, 2018 |
Jun 3, 2016 | I am planning to attach 1760 pin BGA. The pitch is 1mm and Balls are SAC alloy type. The PCB is for
Aug 19, 2015 | Looking for a Santa Clara area contract service with capability to attach 15mil dia. solder coated C
Aug 10, 2015 | Thanks for your reply, Rob. I wish I could attach images here, but despite the option for 'attachme
Aug 5, 2015 | could u please attach the snapshot for the same
Feb 13, 2015 | What would be a recommended way and were to attach therms for profiling on a board with a BGA or BGA
Aug 1, 2014 | Check attachments.
Apr 15, 2014 | Can you attach a photo of said component? Reese
Nov 12, 2013 | Dave, See attachments. Thanks.
Aug 28, 2013 | Thanks, now the attachment link works.
Aug 27, 2013 | When I clicked on "attachment" a new page with a box for adding attachments appeared. Try again???
Jul 11, 2013 | Not sure if this will help or not, but I attached a couple of files that I had laying around.
Jul 9, 2013 | formula(s) attached.
Apr 25, 2013 | Hi, I attach two pictures for you. Hope they help
Apr 11, 2013 | Hi All, sorry to bug you, the attached picture of a LED is driving me nuts,
Oct 23, 2012 | I have attached a PDF overview of some test we have done and the results. (fixed oven profile)
Dec 12, 2011 |
Mar 29, 2011 | Hi 1st, Please find attached BGA manual this may help you, regards llew
Feb 28, 2011 | Davef I tried to click the "Attachment" button, but it trigger "Post" button, so I can't attach the
Nov 3, 2010 | Hi, Can someone recommend an adhesive / epoxy for attaching the tips of thermocouples to the PCB?
Apr 9, 2010 | Accuracy for a 4791 is +- .004. I have attached the GS.
Mar 25, 2010 |
Mar 24, 2010 |
Mar 23, 2010 |
Dec 24, 2009 | http://shanelo.co.za/how_to_attach_a_thermocouple.htm
Dec 6, 2009 | Attached is the BGA BALL soldered underneath reflow profile , plz check and advice.
Oct 26, 2009 | pls find attached photo from Supplier saying that the failure at pth hole was due to blistering caus
Jul 17, 2009 | See attachment. Hope it is useful.
Jul 2, 2009 | On first glance it looks like those particular pads were not attached to any traces. If I had to gue
Jul 1, 2009 | can anyone suggest possible causes or further analysis for the lifted pads in attached picture. than
Jun 12, 2009 | Minimize scavanging by the attached trace and provide more uniform temperatures between all pads.
Jun 5, 2009 | Look in attachment what NASA has to say about lead free electronics
May 27, 2009 |
May 27, 2009 | Please describe the break in the components. Attaching pictures would be helpful.
May 5, 2009 | Check out the attached and compare your issue w/ the issue in the attached. If it's 'beading' versu
Apr 7, 2009 | Board delaminated after reflow - see attached picture. Does moisture cause this problem?
Nov 18, 2008 | See the attached PDF
Nov 6, 2008 |
Oct 31, 2008 | The following attachment I hope is useful.
Sep 18, 2008 | Hi Please see attachment Adam
Sep 16, 2008 | Would you attach pix of the sections?
Aug 6, 2008 | see attachment
Jul 30, 2008 | Ours are 1996 models. No special hoses needed, just attach the air hose and your ready to run.
May 23, 2008 | see attachment (no further comments)
Mar 4, 2008 | The attached photo from cross section. (like head in pillow)
Jan 23, 2008 | Se�or Tech: There's check-box at the bottom of the attachment screen
Sep 15, 2007 | Hi, I attach the termocouples with high temperature solder wire, at the QFP pins.Other thermocopl
Sep 10, 2007 | How are you attaching your thermocouples to measure the temperature of the lead/pad connection?